Are you referring to the strangely high maximum temperature in the y axis? That looks like the ADC read a high transient value. This usually happens while plugging in or unplugging a probe. Since there is no extra high point on the screen, I would guess it happened when unplugging channel four.
I believe the most recent firmware in github has scali g implemented and would fix that issue.